- Registered on: 27/06/2019
- Last connection: 09/09/2020
- 01:36 PM INCLINE - ICP Bug #3424 (Resolved): Mains water supply interruption
- 11am: switched pump controller on/off.
Allowed controller to heat up to ~ 60 °C.
Switched on. Turbo OK.
- 10:11 AM INCLINE - ICP Bug #3424 (Resolved): Mains water supply interruption
- The mains water supply was interrupted during the morning (6 am ish).
Primary pump overheated, so shut down.
- 04:03 PM Vacuum Technology for LTP experiments Updated: Pfeiffer vacuum-technology-book-ii-part-2
- 01:07 PM INCLINE - ICP Updated: iGX Dry Pumping Systems Instruction Manual
- 01:07 PM INCLINE - ICP Updated: Brooks 0250-series MFC controller manual
- 01:07 PM INCLINE - ICP Updated: BOC EDWARDS M150 gas reactor column manual
- 01:07 PM INCLINE - ICP Updated: Advanced Energy RFX600 generator manual
- 01:05 PM INCLINE - ICP Downloaded: Brooks 5850TR-MFC manual
- 01:02 PM INCLINE - ICP Task #3421: Repair Cl2 mass flow controller
- Subject: RE: reparation 5850TR
Date: Thu, 19 Sep 2019 11:31:42 +0200
- 11:40 AM INCLINE - ICP Task #3421 (New): Repair Cl2 mass flow controller
- Defective Brooks Model 5850TR
Also available in: Atom